Level indicators wholesale manufacturer China? The interference comes not only from the outside, but also from the inside of the radar level gauge, such as interference caused by wires, inductance and capacitance between the power transformer and electronic components. In addition, the internal components can also generate noise interference. Today, most radar level gauges have also begun to be improved, using high-frequency microwave technology, which greatly improves the performance of the level gauge and reduces interference. See more details at level indicator.
Level Measurement Considerations: Taking advantage of a specific technology’s ability to reliably address the level measurement in either of these vessels, especially the blowdown flash tank, in a plug-and-play type installation and commissioning format is an easy way to ensure optimal performance. This forgoes calibration, external hardware or inputs. Estimates of up to 49 percent of the energy can be recovered through the use of flash steam routed to heat exchangers or the deaerator to preheat boiler makeup water or support the deaeration process, respectively. In addition, better level control technology at the boiler side eliminates energy losses resulting from unnecessary blowdown to prevent carryover conditions.
Secondly, in cement production, material level measurement encountered another difficulty is strong dust interference, especially pneumatic conveying powder silo, dust flying when feeding, low visibility, laser level meter and high energy ultrasonic level meter can not be measured, although radar level meter at this time can receive part of the surface echo. But the echo signal was also weakened. In addition, due to the uneven surface of the radar echo is refracted, will also lead to the existence of radar echo; In addition, there are some conditions from the bottom to the bin inflation, so that the material surface loose, material level measurement is more difficult.
Measurement accuracy, the accuracy of ordinary radar is generally ±10mm, and the accuracy of precision radar is ±3mm. Selection according to the actual needs of production. Range, according to actual needs, choose the antenna size. Note that the actual range is reduced in complex environments. Antenna type and antenna size, the larger the antenna size, the larger the measured range and the stronger the anti-interference ability. The antenna types of radar level gauge are rod type, bell mouth type, paraboloid and so on.
The electromagnetic wave emitted during measurement can pass through the vacuum, no transmission medium is required, and the anti-interference ability is strong, and it is not affected by temperature, wind, water vapor, water mist, rain, etc. It can be used for liquid level measurement of almost all liquids, and can be continuously measured on-line unattended. Large measurement range, high precision, mature technology and stable performance.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read additional info on https://www.kaidi86.com/. OEM Service-can be customized according to the application of customer,such as level gauge,level switch and level indicator.
There is AC interference and the voltage is high. For example, for the radar level meter used in the production line, the power supply requirement is 24VDC (typical value), but in the on-site measurement, it is found that the power supply is displayed as 27.2V, which is significantly higher than 24VDC, resulting in a large measurement result and even a radar level meter. crash phenomenon. The installation position of the radar level meter is incorrect, which leads to deviations in the measurement. For example, the accumulation of aggregates in the transfer bin is a “mountain”-shaped cone, but only one radar level meter is installed near the discharge port of the return belt. , the installation position is too close to the discharge opening of the return belt, and at the same time, it is too far from the discharge opening of the feeding belt on both sides. Just below the radar level meter is the drop point of the return belt. If the distance is too close, the aggregate in the falling process will interfere with the radar level meter and form false reflections.
The installation of the liquid level gauge should be away from the inlet, outlet, eddy current or protrusions on the inner wall to avoid the interference of electromagnetic waves and affect the measurement. When installing the guided wave radar, make sure that the diameter of the short pipe at the flange connection is greater than the height of the short pipe, otherwise the cable will contact the short pipe wall due to shaking, and the liquid level cannot be measured accurately. How to choose a model, in the final analysis, you still need to understand your own working conditions and the characteristics of different instruments, and judge whether the two match, so that you can get twice the result with half the effort.Want to know about ultrasonic flow meter advantages and disadvantages, click here.